CCD Based Constant Deviation Spectrometer

(For Arc Emission & Absorption Measurements)

Model: HO-ED-S-0203

The apparatus uses CCD imaging instead of photographic silver halide films. It has become difficult to carry out silver halide film based spectroscopic experiments in class rooms due to the unavailability of consumables and hurdles in its processing. CCD imaging of the spectrum is a timely solution. Data acquisition is easy and fast. As most of the commercial spectroscopes are based on digital imaging, students get an exposure to the latest trends in spectroscopy.

Mercury is used as a standard element in most cases. Knowing the wavelength, corresponding to different spectral lines, elements can be judged. Hartmann's method is employed to determine the unknown wavelength.

If a transparent substance is placed between a continuous source and a spectrometer, the substance absorbs a certain part of the spectrum. Hence in the continuous spectrum of the source, a few dark lines or bands appear. This spectrum is called absorption spectrum of the substance. This method is useful for manufacturing as well as for academic experiments. Constant Deviation Spectrometer (Model No: HO-ED-S-0203) can be used for the study of spectral series viz. emission spectra of elements, absorption spectra of compounds, quantitative spectrum analysis etc.

When a substance is heated, electrons first absorb energy, go to a higher energy level and eventually fall to a lower level, by emitting photons. Every element emits a characteristic spectrum of its own. There are certain important and prominent spectral lines associated with each element. These lines are used to distinguish one element from another. In the experiment to determine an element, spectrum of standard known source is pictured first.

Experiment Examples

To measure the wavelength of absorption bands of KMnO4 and calculate it’s Hartmann's constant.

The positions of spectral lines are usually measured in the direction of increasing wavelength. The constant d depends on scale setting, C on the range of spectrum and is a constant for the instrument. To determine these constant, we take three prominent lines of standard spectrum and equations are,

A    =    (λ2 – λ1) / (λ3 – λ1)

B    =    (d2 – d1) / (d3 – d1)

C    =    [ (λ2- λ1) (d0 – d1) (d02) ] / (d2 - d1)

d0    =    (B d3 - A d2) / (B - A)

λ0    =    (Aλ3 - B λ2) / (A- B)

Where, A, B, C, λ0, d0 are called Hartmann’s constants.
Using these Hartmann’s constants, we can calculate the absorption band wavelength as

λ    =    λ0 + [C / (d0 - d)]

To find wavelength of prominent lines of the emission spectra of copper, iron and brass

The wave length of the corresponding lines of the arc spectrum can be calculated using the equation;

λ    =    λ0 + [C / (d0 - d)]

Specifications

Design : Constant deviation 120 F

Prism type : Pellin Broca (N-SF1)

Refractive index : 1.7

Wavelength range : 380 - 680nm

Resolution : 0.1 nm

Input : Micrometer controlled slit
Camera Specifications

CCD sensor : line sensor

Number of pixels : 3648

Pixel size : 800 x 200 micron

Exposure time range : 0.1ms - 6,500ms

Frame rate : Upto 138 scans / second

Interface : USB 2.0

Compatibility : Windows 7 or higher
Features

Modern, user friendly and precision design

Emission and absorption spectra can be observed

CCD imaging system

High-quality optics produce sharp and clear spectra
Scope of Supply

Constant Deviation Spectrometer

Model No: ED-S-0203-CDS
Quantity
:
1 no.

(Specification is as per above)

Metal Arc Stand

Model No: ED-S-0203-MAS
Substance Holder
:
Ceramic
Arc Cavity
:
0 - 25 mm
Quantity
:
1 no.

Copper, Iron, Brass Rods

Model No: ED-S-0203-CIBR
Length
:
55 mm
Diameter
:
8 mm
Quantity
:
10 nos. Each

Metal Arc Power supply

Model No: ED-S-0203-MAPS
Input
:
230V, 50 Hz
Output
:
0 - 100V, 0 - 10A
Quantity
:
1 no.

Mercury Vapor Lamp with Power Supply

Model No: ED-S-0203-MLPS
Input
:
230V, 50 Hz
Output
:
4KV, 30mA
Quantity
:
1 no.

Incandescent Lamp

Model No: ED-S-0203-IL
Input
:
230V, 50 Hz
Output
:
25 Watts
Quantity
:
1 no.

CCD Camera

Model No: ED-S-0203-CCDC
Resolution
:
Selectable (320 - 1280 pixels)
Connectivity
:
USB 2.0
Quantity
:
1 no.

Accessories

Cuvette
KMnO4 Powder
Grinding File
Software CD
Instruction Manual
Net Book (Optional)